Spectrometer Design Guide
This spectrometer design guide provides some simple and easy to use design guidelines and formulas for designing, evaluating and comparing various diode array, diffraction grating based spectrometers designs.
The input to the design process is the wavelength range, you want to cover, and the optical resolution, by which you need to resolve the various structures in your spectrum (often peaks).
How a diode detector array spectrometer works
Basically, a spectrometer is an optical system consisting of two lenses/mirrors that produces an image of the input slit on the detector. In between the lenses/mirrors is placed a diffraction grating which disperses different wavelengths in different angles. This causes different wavelengths of light entering the input slit to be imaged to different position on the detector array.
Back to the top
Transmission grating based spectrometer
Back to the top
Flow chartBack to the top
The Spectrometer Guides 8 stepsStep 1: Choose geometry
The first step is to choose among the Czerny-Turner or LGL type geometries. For the Czerny Turner a typical value for Θ is around 30° wheras transmission gratings are generally used in the Littrow configuration and –1st order where α = -β => Θ = 0°.
Step 2: Choose grating
The second step is to choose a diffraction grating. Most grating vendors have an on-line catalogue where you can find one or more grating options to try in your design. You should choose a grating that has high diffraction efficiency in your wavelength. The important parameter that you shall use for the design in the next steps is the groove density G.
Step 3: Calculate diffraction angle
The angle of incidence α on the grating and the diffraction angle β for the center wavelength λc are key parameters in the spectrometer design. These angles can be calculated once the grating groove density G and the total deflection Θ is chosen.
Step 4: Choose detector
The purpose of the spectrometer design is to disperse the wavelength range across the width of the detector array LD. There are a large range of diode array detectors specifically designed for spectrometers. In general, if you need a compact spectrometer you should aim for a short detector (typically 1/4” or 6.4 mm). However, if you require a broad spectral range and/or a high resolution you should aim for a wide detector (typically 1/1” or 25.8 mm).
Step 5: Calculate focal length of focus lens
Once the width of the detector is know you can calculate the focal length of the focusing mirror/lens.
Step 6: Choose magnification
As mentioned earlier, the spectrometer is imaging the input slit to the detector and we generally want to have the slit as wide as possible to collect as much light through the input slit as possible. Therefore, the magnification in the system Μ should preferably be close to 1 which means that the width of the input slit ideally is imaged 1:1 onto the detector array.
Step 7: Calculate focal length of colli lens
As in any imaging system the magnification is determined by the ratio between the focal lengths of the two lenses in the system. For a spectrometer this ratio has to be slightly modified due to the deflection along the beam path in the grating. However, once the magnification is chosen the focal length of the collimation mirror/lens can easily be calculated.
Step 8: Calculate input slit width
The input slit width ωslit is determined by the required optical resolution Dl and the magnification. Once you know your input slit width you are ready to evaluate if your spectrometer design is viable or you have to go back and change some of your choices for grating, detector, or magnification for instance.
Is the input slit width practical?Input slits comes in widths down to 5 microns but such narrow slits will only allow very limited amount of light to enter your spectrometer. So, if your design requires a slit of 5 – 10 microns or less you could consider the following:
- loosening your requirement for the resolution
- choosing a wider detector and choosing a grating with a higher groove density
If this value is larger than your required Δλ, your system is diffraction limited by the optics and you will not be able to obtain a better resolution than ΔλdiffractionBack to the top
If this value is larger than your required Δλ, your system is diffraction limited by the optics and you will not be able to obtain a better resolution than ΔλdiffractionSpectrometer Design Guide
Back to the top