Phase Mask Cleaning
Recommended procedure for regular Phase mask cleaning
Even with proper handling, all precision optical components will need cleaning from time to time. Adhering to proper cleaning procedures will lengthen the lifetime of the component. The complex surface structure of Phase masks places special requirements on the nature of the cleaning process.
Routine Cleaning Procedure
Use the following gentle cleaning procedure to remove grease and loose particles:
- Agitate the Phase Mask in a Petri dish filled with Semiconductor Grade Acetone. Ensure that the acetone does not dry out before step 2.
- Agitate the Phase Mask in a Petri dish filled with Semiconductor Grade Isopropanol Alcohol or Semiconductor Grade Ethanol. Ensure again that the fluid does not dry out before step 3.
- Agitate the Phase Mask in a Petri dish filled with ultra-pure water.
- If available, spin drying is optimal. Otherwise, blow dry (preferably in a cleanroom environment) with clean air (maximum pressure 2 bar).
Special cleaning procedure
If a more effective cleaning process is required, Ibsen offers a cleaning service, which includes a proprietary cleaning process. Contact us for more information on this service.